Nikon Photomic FTn Finder, Lightmeter
On Monday, August 31, 1998 at 2:00:00 AM UTC-5, Drwyn wrote:
Repairing the Nikon Ftn is not that difficult. There are two different
problems with these prism meters. One is that the ring resistor gets dirty and
causes the needle to be irratic. The second cause is that the Cds cells change
their resistive properties after long periods of time. I have repaired several
Nikon Ftns by cleaning the ring resistor and replacing the Cds cells. The
meter then must be recalibrated for the new Cds cells which involves measuring
the resistance of the Cds cells for each stop and matching this with the ring
The same goes with the other Nikon meter prisms including the T, Tn Ftn, DP-1
If you want your meter repaired, I would be happy to do it.
I would be happy to send you one. Interested?